For either subnet mask, numbers between 1 and are valid choices for the last number of the IP address. Exit when you are finished using the Ethernet Device Configuration utility. The best way to achieve this is by using instrument drivers. If your network uses DHCP, the network configuration is performed automatically at startup. Install Instrument Drivers Next. If there is a problem, you no longer see your device.
|Date Added:||22 September 2005|
|File Size:||52.57 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The basic functions and concepts described above may seem simple; however, they can be invaluable in troubleshooting and neet your GPIB system up and running.
With the VISAIC utility, you can speed up application development by learning how to automate measurements with your instruments, uncover GPIB problems, and avoid headaches by identifying malfunctioning instruments. You also can use the Resource to Find field to query each instrument for an identification ID string.
The utility displays four options for resolving the verification error. Once you have determined the GPIB address of your instruments, it is easy to establish communications to verify that you can send and receive data to and from the instrument.
You have just confirmed communication with your instrument. ID querying is discussed in the next section. Refresh the list of devices, after the device reboots, to verify that the hostname is now valid.
Instruments must be powered on and connected to the GPIB controller to be recognized. If it is not, then you know to check cables, confirm power, and verify that the instrument is working properly.
All devices on an Ethernet network are assigned a unique physical address—the Ethernet address—so they can communicate with each other. If you are assembling your own small Ethernet network, you can choose your own IP addresses.
National Instruments GPIB ENET 100 Ethernet to GPIB Controller
For either subnet mask, numbers between 1 and are valid choices for the last number of the IP address. Connect one end of the power cord to the power supply.
Then select your GPIB controller. If you select Obtain an IP address automatically DHCPyou do not need to enter any network parameters unless you want to change the hostname of the Ethernet device. Also have the network administrator assign the proper subnet mask, gateway, and DNS server addresses. It makes GPIB instrument detection and control easy by providing tools that help you ennet for connecting instruments, and send and receive communication with your device.
Installing GPIB-ENET/ Controllers – National Instruments
This is indicated by an exclamation point! Consult your network administrator if you do not know whether your network uses DHCP.
Connect the other end to your GPIB instrument. Install Instrument Drivers Next.
National Instruments GPIB-ENET / GPIB-ENET/ – In Stock, We Buy Sell Repair, Price Quote
The Ethernet address is not the IP address. Run the Ethernet Device Configuration Utility. The best way to achieve this is by using instrument drivers.
Numbers between 0 and are valid for the third number of the IP address, but this number must be the same as other devices on your network if your subnet mask is This indicates that you should check for loose or disconnected cables, power disruption, or a malfunctioning instrument.
Select the one that best fits your situation. Gplb the eney end of the Ethernet cable to your Ethernet network. The unit is now ready to operate.
If your network uses DHCP, the network configuration is performed automatically at startup. If you have two or more instruments on the bus, you can disconnect all instruments except one to determine its address.
Alternatively, system power may be disrupted for a variety of reasons or the actual instrument may begin to malfunction.
Unfortunately, engineers sometimes forget to reconnect all the cables and power up all the instruments.